Surgery instruments like a safety scalpel must be kept in proper shape as they are vital towards carrying out an operation properly in USA. Surgical instruments are designed to ensure that the surgeons can complete surgeries in an effective and safe manner using safety scalpels and other tools. Surgeries are mainly held to modify or remove any kind of body tissues. Surgical instruments also come in a variety of types designed to do separate things. They are categorized based on their usage. Also they are sometimes categorized according to the materials they are made of.
Disposable and non disposable surgical scalpels:
The scalpel is one of the most common surgical instruments, and recently safety scalpels in USA are the most used of them. These are used for incisions or thin cuts on the flesh to reach the inner organs. The retractors are used used to keep the cuts open so that the surgeons can reach in and go through with the operation. Forceps are another popular surgical instrument which is used for picking and holding tissues. When it comes to cutting tissues scissors are also used but these scissors are not the same as the household ones. These are much more precise and effective. All these instruments are made from hardened and tampered steel. After considering their composition they are labelled as reusable or not. There are also other surgical instruments which are made for just one time usage. These are destroyed after usage like the disposable surgical needles. Disposable surgical scalpels are always sterilized the moment they are purchased in home and hospital departmental stores in USA. These disposable tools are generally made from runner or plastic materials. Protective surgical gloves, twin bone nasal catheter, nasal oxygen catheter, oxygen mask, suction catheter, and trachea’s tony tubes are few examples of these types of instruments.
Laprascopy With the Harmonic Scalpel34 surgical staples closing scalp following craniotomy
Surgical staples are specialized staples used in surgery in place of sutures to close skin wounds, connect or remove parts of the bowels or lungs. A more recent development, from the 1990s, uses clips instead of staples for some applications; this does not require the staple to penetrate.
Stapling is much faster than suturing by hand, and also more accurate and consistent. Staples are primarily used in bowel and lung surgery, because staple lines are more consistent and therefore less likely to leak blood, air or bowel contents. Still, several randomized controlled trials have shown no significant difference in bowel leakage after anastomoses performed either manually with suture by experienced surgeons, or after mechanical anastomoses with staples. In skin closure, dermal adhesives (skin glues) are also an increasingly common alternative.
Staplers were originally developed to address the perceived problem of patency (security against stenosis or occlusion of the lumen) and tightness (security against leaks of blood or bowel contents) as well as easiness and quickness in performing the anastomosis. Leaks from poor suturing of bowel anastomoses was at that time a significant cause of post-surgical mortality. More recent studies have shown that with current suturing techniques there is no significant difference in outcome between hand sutured and mechanical anastomoses, but mechanical anastomoses are significantly quicker to perform.
The technique was pioneered by a Hungarian surgeon, Humor Hultl, known as the "father of surgical stapling". Hultl's prototype stapler of 1908 weighed eight pounds (3.6 kg), and required two hours to assemble and load. Many hours were spent trying to achieve a consistent staple line and reliably patent anastomoses.
The early instruments, by developers including Hultl, von Petz, Friedrich and Nakayama, were complex and cumbersome to use. The technology was refined in the 1950s in the Soviet Union, allowing for the first commercially produced re-usable stapling devices for creation of bowel and vascular anastomoses. Mark M. Ravitch, brought a sample of stapling device after attending a surgical conference in USSR, and introduced it to entrepreneur Leon C. Hirsch, who founded the United States Surgical Corporation in 1964 to manufacture surgical staplers under its Auto Suture brand. Until the late 1970s USSC had the market essentially to itself, but in 1977 Johnson & Johnson's Ethicon brand entered the market and today both are widely used, along with competitors from the Far East. USSC was bought by Tyco Healthcare in 1998, which became Covidien on June 29, 2007.
Safety and patency of mechanical (stapled) bowel anastomoses has been widely studied. It is generally the case in such studies that sutured anastomoses are either comparable or less prone to leakage. It is possible that this is the result of recent advances in suture technology, along with increasingly risk-conscious surgical practice. Certainly modern synthetic sutures are more predictable and less prone to infection than catgut, silk and linen, which were the main suture materials used up to the 1990s.
One key feature of intestinal staplers is that the edges of the stapler act as a haemostat, compressing the edges of the wound and closing blood vessels during the stapling process.Laparoscopic cholecystectomy. Close-up demonstration of a surgical skin stapler.
The first commercial staplers were made of stainless steel with titanium staples loaded into reloadable staple cartridges.
Modern surgical staplers are either disposable and made of plastic, or reusable and made of stainless steel. Both types are generally loaded using disposable cartridges.
The staple line may be straight, curved or circular. Circular staplers are used for end-to-end anastomosis after bowel resection or, somewhat more controversially, in esophagogastric surgery. The instruments may be used in either open or laparoscopic surgery, different instruments are used for each application. Laparoscopic staplers are longer, thinner, and may be articulated to allow for access from a restricted number of trocar ports.
Some staplers incorporate a knife, to complete excision and anastomosis in a single operation. Staplers are used to close both internal and skin wounds. Skin staples are usually applied using a disposable stapler, and removed with a specialized staple remover. Staplers are also used in vertical banded gastroplasty surgery (popularly known as "stomach stapling").Vascular stapler for reducing warm ischemia in organ transplantation. With this model each stapler end can be mounted on donor and recipient by independent surgical teams without care for reciprocal orientation, being the maximal possible vascular axis torsion ≤30°. Activating guide-wire is connected just immediately before firing (video)
While devices for circular end-to-end anastomosis of digestive tract are widely used, in spite of intensive research circular staplers for vascular anastomosis never had yet significant impact on standard hand (Carrel) suture technique. Apart from the different modality of coupling of vascular (everted) in respect to digestive (inverted) stumps, the main basic reason could be that, particularly for small vessels, the manuality and precision required just for positioning on vascular stumps and actioning any device cannot be significantly inferior to that required to carry out the standard hand suture, then making of little utility the use of any device. An exception to that however could be organ transplantation where these two phases, i.e.device positioning at the vascular stumps and device actioning, can be carried out in different time, by different surgical team, in safe conditions when the time required does not influence donor organ preservation, i.e. at the back table in cold ischemia condition for the donor organ and after native organ removal in the recipient. This is finalized to make as brief as possible the donor organ dangerous warm ischemia phase that can be contained in the couple of minutes or less necessary just to connect the device's ends and actioning the stapler.
Although most surgical staples are made of titanium, stainless steel is more often used in some skin staples and clips. Titanium produces less reaction with the immune system and, being non-ferrous, does not interfere significantly with MRI scanners, although some imaging artifacts may result. Synthetic absorbable (bioabsorbable) staples are also now becoming available, based on polyglycolic acid, as with many synthetic absorbable sutures.
Titanium staples are not suspected of causing nickel reactions because nickel is rarely if ever used in titanium alloys.
Where skin staples are used to seal a skin wound it will be necessary to remove the staples after an appropriate healing period, usually between 5 and 10 days, depending on the location of the wound and other factors. The skin staple remover is a small manual device which consists of a shoe or plate that is sufficiently narrow and thin to insert under the skin staple. The active part is a small blade that when hand-pressure is exerted it pushes down on the staple and pushes it through a slot in the shoe and deforms the staple into an 'M' shape to facilitate its removal, although in an emergency it is possible to remove them with a pair of artery forceps. Skin staple removers are manufactured in many shapes and forms, some disposable and some reusable.
Use of a Laser Scalpel
Electrosurgery is the application of a high-frequency (radio frequency) alternating polarity, electrical current to biological tissue as a means to cut, coagulate, desiccate, or fulgurate tissue. Its benefits include the ability to make precise cuts with limited blood loss. Electrosurgical devices are frequently used during surgical operations helping to prevent blood loss in hospital operating rooms or in outpatient procedures.
In electrosurgical procedures, the tissue is heated by an electric current. Although electrical devices that create a heated probe may be used for the cauterization of tissue in some applications, electrosurgery is usually used to refer to a quite different method than electrocautery. Electrocautery uses heat conduction from a probe heated to a glowing temperature by a direct current (much in the manner of a soldering iron). This may be accomplished by direct current from dry-cells in a penlight-type device.
Electrosurgery, by contrast, uses radio frequency (RF) alternating current to heat the tissue by RF induced intracellular oscillation of ionized molecules that result in an elevation of intracellular temperature. When the intracellular temperature reaches 60 degrees C, instantaneous cell death occurs. If tissue is heated to 60-99 degrees C, the simultaneous processes of tissue desiccation (dehydration) and protein coagulation occur. If the intracellular temperature rapidly reaches 100 degrees C, the intracellular contents undergo a liquid to gas conversion, massive volumetric expansion, and resulting explosive vaporization.
Appropriately applied with electro surgical forceps, desiccation and coagulation result in the occlusion of blood vessels and halting of bleeding. While the process is technically a process of electrocoagulation, the term "electrocautery" is sometimes loosely, nontechnically and incorrectly used to describe it. The process of vaporization can be used to ablate tissue targets, or, by linear extension, used to transect or cut tissue. While the processes of vaporization/ cutting and desiccation/coagulation are best accomplished with relatively low voltage, continuous or near continuous waveforms, the process of fulguration is performed with relatively high voltage modulated waveforms. Fulguration is a superficial type of coagulation, typically created by arcing modulated high voltage current to tissue that is rapidly desiccated and coagulated. The continued application of current to this highly impedant tissue results in resistive heating and the achievement of very high temperatures - enough to cause breakdown of the organic molecules to sugars and even carbon, thus the dark textures from carbonization of tissue.
Diathermy is used by some as a synonym for electrosurgery but in other contexts diathermy means dielectric heating, produced by rotation of molecular dipoles in a high frequency electromagnetic field. This effect is most widely used in microwave ovens or some tissue ablative devices which operate at gigahertz frequencies. Lower frequencies, allowing for deeper penetration, are used in industrial processes.
RF Electrosurgery is commonly used in virtually all surgical disciplines including dermatological, gynecological, cardiac, plastic, ocular, spine, ENT, maxillofacial, orthopedic, urological, neuro- and general surgical procedures as well as certain dental procedures.
RF Electrosurgery is performed using a RF electrosurgical generator (also referred to as an electrosurgical unit or ESU) and a handpiece including one or two electrodes - a monopolar or bipolar instrument. All RF electrosurgery is bipolar so the difference between monopolar and bipolar instruments is that monopolar instruments comprise only one electrode while bipolar instruments include both electrodes in their design.
The monopolar instrument called an "active electrode" when energized, requires the application of another monopolar instrument called a "dispersive electrode" elsewhere on the patient's body that functions to 'defocus' or disperse the RF current thereby preventing thermal injury to the underlying tissue. This dispersive electrode is frequently and mistakenly called a "ground pad" or "neutral electrode". However virtually all currently available RF electrosurgical systems are designed to function with isolated circuits - the dispersive electrode is directly attached to the ESU, not to "ground". The same electrical current is transmitted across both the dispersive electrode and the active electrode, so it is not "neutral". The term "return electrode" is also technically incorrect since alternating electrical currents refer to alternating polarity, a circumstance that results in bidirectional flow across both electrodes in the circuit.
Bipolar instruments generally are designed with two "active" electrodes, such as a forceps for sealing blood vessels. However, the bipolar instrument can be designed such that one electrode is dispersive. The main advantage of bipolar instruments is that the only part of the patient included in the circuit is that which is between the two electrodes, a circumstance that eliminates the risk of current diversion and related adverse events. However, except for those devices designed to function in fluid, it is difficult to vaporize or cut tissue with bipolar instruments.
Neural and muscle cells are electrically-excitable, i.e. they can be stimulated by electric current. In human patients such stimulation may cause acute pain, muscle spasms, and even cardiac arrest. Sensitivity of the nerve and muscle cells to electric field is due to the voltage-gated ion channels present in their cell membranes. Stimulation threshold does not vary much at low frequencies (so called rheobase-constant level). However, the threshold starts increasing with decreasing duration of a pulse (or a cycle) when it drops below a characteristic minimum (so called chronaxie). Typically, chronaxie of neural cells is in the range of 0.1–10 ms, so the sensitivity to electrical stimulation (inverse of the stimulation threshold) decreases with increasing frequency in the kHz range and above. (Note that frequency of the alternating electric current is an inverse of the duration of a single cycle). To minimize the effects of muscle and neural stimulation, electrosurgical equipment typically operates in the radio frequency (RF) range of 100 kHz to 5 MHz.
Operation at higher frequencies also helps minimizing the amount of hydrogen and oxygen generated by electrolysis of water. This is especially important consideration for applications in liquid medium in closed compartments, where generation of gas bubbles may interfere with the procedure. For example, bubbles produced during an operation inside an eye may obscure a field of view.
There are several commonly used electrode configurations or circuit topologies:
With "bipolar" instruments the current is applied to the patient using a pair of similarly-sized electrodes. For example, special forceps, with one tine connected to one pole of the RF generator and the other tine connected to the other pole of the generator. When a piece of tissue is held by the forceps, the RF alternating polarity electrical current oscillates between the two forceps tines, heating the intervening tissue by the previously described synchronous oscillation of intracellular ions.
In monopolar configuration the patient is attached to the dispersive electrode, a relatively large metal plate or a flexible metalized plastic pad which is connected to the RF generator or electrosurgical unit (ESU). The surgeon uses a pointed or blade shaped electrode called the "active electrode" to make contact with the tissue and exert a tissue effect...vaporization, and it's linear propagation called electrosurgical cutting, or the combination of desiccation and protein coagulation used to seal blood vessels for the purpose of Hemostasis. The electric current oscillates between the active electrode and the dispersive electrode with the entire patient interposed between the two. Since the concentration of the RF current reduces with distance from the active electrode the current density rapidly (quadratically) decreases. Since the rate of tissue heating is proportional to the square of current density, the heating occurs in a very localized region, only near the portion of the electrode, usually the tip, near to or in contact with the target tissue.
On an extremity such as a finger, there is limited cross-sectional area to disperse the current, a circumstance which might result in higher current density and some heating throughout the volume of the extremity.
Another bipolar instrument is characterized with both electrodes on the same design, but the dispersive electrode is much larger than the active one. Since current density is higher in front of the smaller electrode, the heating and associated tissue effects take place only (or primarily) in front of the active electrode, and exact position of the dispersive electrode on tissue is not critical.
- USA stainless steel blade
- Newyork scalpel handle use
- Los Angeles laryngoscope blades
- Chicago microtome blades
- Dallas surgical blade
- Houston no 11 surgical blade
- Washington microtome blades
- Philadelphia ultrasonic scalpel
- Miami dermaplaning blades
- Atlanta dissecting scissors
- Boston surgical blade
- San Francisco microtome blades
- Phoenix laryngoscope handle
- Detroit swann morton scalpel handle
- Seattle disposable blades
- San Bernardino ophthalmic surgical instruments
- San Diego microtome blades
- Denver microtome blades
- Charlotte surgical instruments
- Cleveland ultrasonic scalpel
- Salt Lake City surgical supplies
- Las Vegas stainless steel blade
- Ogden Clearfield ophthalmic surgical instruments
- Provo Orem ultrasonic scalpel
- St George surgical blade
- San Antonio surgical supplies
- Austin no 11 surgical blade
- Jacksonville scalpel handle use
- Columbus surgical blade
- Indianapolis laryngoscope blades
- Fort Worth surgical instruments
- El Paso no 11 surgical blade
- Nashville ophthalmic surgical instruments
- Memphis microtome blades
- Portland surgical supplies
- Oklahoma City dermaplaning blades
- Louisville dissecting scissors
- Baltimore microtome blades
- Milwaukee dissecting scissors
- Albuquerque microtome blades
- Tucson surgical blade
- Fresno swann morton scalpel handle
- Sacramento swann morton scalpel handle
- Mesa swann morton scalpel handle
- Kansas City ultrasonic scalpel
- Long Beach scalpel handle use
- Colorado Springs scalpel handle use
- Raleigh surgical instruments
- Virginia Beach surgical instruments